# Uncertainty of measurements – How to quantify ambiguity of your results?

Ambiguity or uncertainty refers to a degree of doubt on the results of testing. It is an accepted fact that it is never possible to report the actual value as there is always a scope for error, however small it may be, that results in variation between the true value and reported value.

Measurement is truly valid if it is accompanied by definition of degree of uncertainty. Such uncertainties arise from deviations from stipulated environmental controls, non-homogeneity of sample, method or instrumental limitations and personal operator errors. Some of these errors are measurable and can be corrected through calibration whereas others are indeterminate. In other words every measurement or test should be defined in terms of two parameters. First is the bandwidth or interval defining the accuracy of the measurement and the second is the level of confidence.

A number of factors contribute to uncertainty of measurements made in a chemical laboratory. Some of these factors are:

• Sample homogeneity
• Sample degradation between collection and analysis
• Contamination during sampling, sample treatment or storage
• Incomplete extraction of analyte from sample matrix
• Interferences arising from other sample matrix components
• Variations in recommended environmental controls
• Personal bias in recording observations
• Errors in weighing or volumetric transfer of solutions
• Degradation of standards and reference materials over storage

#### Expression of uncertainty

A set of repeat observations should be made and both mean and standard deviation (SD) are reported. From these the value of uncertainty, U, is calculated as

$$U =\frac{SD}\sqrt{n}$$

where n is the number of observations.

The combined uncertainty resulting from several factors is expressed as

Combined uncertainty,

$$U_C = \sqrt{a^2 + b^2 + c^2 + d^2+……}$$

where a, b, c, etc are individual uncertainties.

Expanded uncertainty, Ue, is arrived at by multiplying the combined uncertainty with the coverage factor, k

$$Ue = kU_c$$

Average factor k of 2 is used to report expanded uncertainty at a confidence level of 95%

Finally the observation is reported to 2 significant digits as

Result = Mean value +/- Ue

In any analysis it is important to appreciate the role of factors that can contribute to uncertainty of measurements. There is a need to identify, minimize and wherever possible quantify them through frequent calibration against reference materials and inter- laboratory proficiency testing.

## A simple way to understand and practice the clauses covered by ISO 17025:2005

I have been a part of an accredited laboratory for 10 years now and have successfully faced more than 12 audits based on the ISO…

## 125 Job Profiles for an Analytical Chemist!

The ever increasing consumer awareness and the demand for quality have made analytical chemistry and analytical chemist, an integral and essential part of all industries.…

## Essentials of Good Laboratory Practices (GLP)

Quality Product testing is the backbone for promotion of global trade. A product tested and approved to meet defined specifications should qualify when tested in…

## Importance of maintaining Laboratory Log Books

A laboratory log book is a tamper proof record of laboratory activities related to use of testing equipments, environmental conditions in laboratory and consumption of…

## Coordination & Strong Communication-Pillars of success of Laboratory Operations

Harmony and cordial relations between individuals and groups play an important role in streamlined and efficient operations in an organization. The same holds true for…