# How to Choose a Gas Chromatographic Detector for my Analysis?

In Gas Chromatography detection is based on a property of the eluting compound which is different from the carrier gas. It is necessary to have a clear understanding of your analysis requirements in advance before you can decide on the suitability of the detector. Important considerations are:

• Is the detector Destructive or Nondestructive. Nondestructive detectors are concentration sensitive and can be used in series before destructive detectors. On the other hand destructive detectors are mass sensing and cannot be used in conjunction with other detectors.
• Sensitivity requirements in terms of lowest detectable quantity and large linear dynamic range.
• Selectivity refers to what type of compounds a detector will respond to falls into two categories
• Universal detectors respond to all compounds reaching them from the column
• Selective or specific detectors respond only to a class of compounds containing a particular functional group (selective) or only to a particular compound (specific)

Once you have the clarity on the requirements of your analysis the next stage is selection of your Gas Chromatographic detector from the available options.

We shall consider some common detection options available:

#### Flame Ionization detector

Flame ionization detector is the most commonly used detector as it responds to most organic compounds that can ionize in the hydrogen – air flame. It has a vast linear dynamic range of $$10^ 7$$.

It is not suitable for inorganic gases such as CO, $$O_2$$, $$CS_2$$, $$N_2$$, $$CO_2$$, $$H_2O$$ and $$NO_x$$ and inert gases.

#### Thermal Conductivity detector

Thermal Conductivity detector is nondestructive and universal. It responds to compounds which have thermal conductivites different from the carrier gas. Inorganic gases which cannot be detected on FID can be analysed with the TCD. Its linear dynamic range is $$10^6$$. It has useful applications in analysis of natural and refinery gases.

#### Electron capture detector

Electron detector is nondestructive and selective to compounds with electro negative substituents such as halogens, nitriles, nitrates, conjugated carbonyls, organometallics and oxygenated components. Linearity is $$10^5$$ and applications include chlorinated pesticides, polyfluorinated biphenyls and herbicides.

#### Nitrogen phosphorus detector

Nitrogen phosphorus detector responds selectively to most compounds that contain nitrogen or phosphorus. Linear dynamic range is $$10^6$$. Widely used for drugs of abuse and pesticides containing nitrogen or phosphorus.

#### Flame Photometric detector

Flame Photometric Gas Chromatographicdetector is destructive and selective. It is sensitive to compounds containing phosphorus or sulphur. Linear dynamic ranges from $$10^3$$ for S to $$10^4$$ for P. It is useful for analysis of organophosphorus pesticides, sulphurf in petroleum products and pulp milling process studies.

#### Photo Ionization detector

Photo Ionization Gas Chromatographic detector is a nondestructive and selective detector. It responds to compounds ionized by UV light such as aromatic and unsaturated compounds in drinking water, wastewater and sludge. Benzene, toluene, xylene and PAH’s in petroleum products. Linear dynamic range is $$10^7$$.

#### Electrolytic Conductivity detector or Hall detector

Electro Conductive the Gas Chromatographic detector is sensitive to halogens, sulphur and nitrogen compounds, PCB’s, volatile organic compounds, sulphur compounds such as petroleum products and pesticides. Linear dynamic range is $$10^6$$ for chlorine.

#### Mass Selective detectors

These Gas Chromatographic detectors are destructive and are based on the fragmentation of molecules by electron bombardment and separation based on mass to charge ratio. Linear dynamic range is $$10^5$$. Applications include pharmaceuticals, environmental monitoring and pesticide residue analysis.

## Selecting a suitable detector for Gas Chromatographic analysis

When it comes to defining Gas chromatography (GC), we all are aware that it is an analytical separation technique that analyses the volatile substances present…

## Analytical Method Validation

The analytical results you communicate can have far-reaching consequences and can form the basis for taking decision on safety of use of commercial products, foods…

## Gas Chromatography in Petroleum Refining Industry

Gas chromatography has found large-scale applications in monitoring of petroleum refining and finished product testing. This is evidenced by evolution of several dedicated versions of…

## What is Chromatography?

If you work in the chemical industry, you must have heard about the technique of chromatography. Despite being a common procedure, many people still aren’t…

## Observing Microorganism

Part 1 – Microscopy Microbiology deals with the study of microorganisms that cannot be seen distinctly with the unaided eye. Observation of microorganisms is an…

• ## Course

+72 enrolled
Not Enrolled

#### Course Includes

• 19 Lessons
• 67 Topics
• 3 Quizzes
• Course Certificate